It''s hard to tell from your description, but at least from the rough picture, I only see one star in the image. Could that be the problem? Does the image look OK to you in person? E.g. wait till darker, use longer exposure or gain, ...? If you capture an image with the same settings and upload it to astrometry.net, does it solve there?
I don't think anything changed with respect to plate solving / polar alignment in this release.
I don't have much experience with such a wide field of view, but as you say, it worked before.
I'm experiencing the same thing. As I mentioned in my post about testing SM 1.6.0, I could not get the internal plate solver to work (I attached the logs). I changed to Astrometry and it all worked fine. I've had no problems with the internal solver until KStars 3.5.5.
Mounts: Sky-Watcher EQ6-R Pro, Meade LX85, Celestron NexStar Evolution Alt/Az
OTAs: Celestron 8" Edge HD w/Celestron Focus Motor, Meade 80mm APO Triplet Refractor w/ZWO EAF
Cameras: ASI533MC Pro, ASI183MC Pro, ASI224MC, ASI120MC-S, ZWO ASI290MM
Raspberry Pi 4 with Stellarmate OS, MacBook Pro
I also had "DISABLED: FOV must be 10 arcmins or wider. 60+arcminutes is recommended." several times after updating to 1.6.0.
Only on one of my two setups, however.
I blamed a flaky USB cable connection.
RPI 4 B (4Gb) with SMate
Nikon D5500 H-a modified. Nikon D5500 stock
Askar FRA400, TPO RC6
ZWO ASI120MM Mini on ZWO 30F4
ZWO ASI224MC-S on Orion 50 mm f/3.2
ZWO EAF x 2
Ron, @Euripides: I'm no expert on plate solving, but am willing to work with you to figure this out.
First, the message "DISABLED: FOV must be 10 arcmins or wider. 60+arcminutes is recommended." is worrysome.
I looked at the code, and that's output when, just as it says, it calculates the FOV too small. It calculates the FOV from this formula:
- fov_x is in arc-minutes,
- ccd_width is the # of pixels along the width of your imager,
- cc_hor_pixel is the width, in microns, of your pixels,
- focal_length in mm.
So, for instance, in my setup with FL=2000, px=3.8, ccd_width=4656 I get fov_x = 30 arc-minutes.
Check the Indi window for your imager to make sure the value is good.
Here's mine, for my ASI1600, correctly set at 3.8 microns and 4656 pixels
Next, assuming there's no issue there, we can go into testing the plate solving itself.
Ekos could definitely use an easy way to debug plate solving, offline, during the daytime, etc.
Here's one way to do it, not as easy as it should be, but a start.
- Capture an image (in the capture tab) with the same exposure/gain/etc as you're using with Align. This will be out test file for further investigation.
- Switch to the simulator
- Jasem introduced an interesting feature in the simulator a while back, where it can replay images from a directory, instead of creating images.
You can find that if you run the simulator profile, and start up Ekos. Go to the indi tab labeled 'CCD Simulator' and inside there go to the tab labelled 'Simulator Config'.
At the bottom you will find ''Use Dir". Enable that. Note the directory location on the above line, and put your test fits file in that directory (and temporarily remove any other files
that might be in that directory--you probably won't have any there). Now, when you run the simulator, it should always "replay" that fits file as the new capture. Remember to uncheck that 'Use dir" button later, or just don't save this configuration.
- For sanity's sake, capture a few images with the simulator. You should always see your test image.
- Go to the align tab, Use 'Capture and Solve' with Action = 'Nothing'. It should try to solve your image, and now you can play with parameters to get it to solve.
Note, you probably will also need to configure the simulator to have the focal length and ccd parameters for your real system where the image was captured.
Enter the CCD stats on that same 'Simulator Config' page as above, and the Focal Length on the mount tab.
I think you're safe that changing these things on the simulator won't affect you when you go back to your "real profile", but please double check.
If you can't get a fits file to solve, please upload it somewhere and let me (and any others) have a crack at it too.
Of course, if you do that, send the relevant image parameters (focal length, pixel size, num pixels).
The warning message is an indicator that something is wrong. Can you go to Image Info tab of the camera INDI Control Panel and check that out? also the mount module, what's the focal length listed there?
The only thing remaining that would be possible is language and different between decimal point and comma, but that'd hardly an issue now as it was addressed long ago. But I'm running out of ideas and this is the only thing that would explains it.