What kind of debugging environment?
Going to guess the alignment module has different approach. Did you try setting a target first then doing the goto?
I think what's happening is that you're not setting the target in Ekos before you slew. If all you do is right click the star before you lx200->goto , then Ekos hasn't yet set the target. Try doing "center and track" after selecting the star to ensure its set as the current target, and then do the goto, and see if that works..
Giving this a little more thought, I can visualize a "predictive focus tool" that could set the correct initial position and step size based on local conditions and the parameters of the rig being measured, by using a feedback loop created by logging the values of the independent variables (that have a significant enough effect to bother with) of each focus attempt, including the available weather data, into a text file database. This would need to be performed against the same starfield for some base number of repetitions at different weather conditions. You'd need to input the measured step distance measured, and the tool would use the telescope parameters configured elsewhere to calculate things like step-width of the CFZ and use the curves generated by the data set at each variable point to build a mesh of actual resulting values, that could be used as a predictor for focus setting(s) that should be pretty accurate once it has enough training runs in it. So as a user, you'd do the measurement, configure the tool with it, and select a reliable and non-variable star (like polaris, for example), run "Train Focuser" (autofocus that saves values to training file). Then, every time you set up, point to Polaris, and run "Train Focuser" again. Do it enough times, and it should get pretty good at predicting focus position for a given set of conditions. The mesh generated, likely might have value elsewhere in the software as well, since it effectively models the performance of your rig at a given location over time.
Let me simplify then. For the telescope you used to generate this graph, calculate the width of the CFZ , and the focal length change of a single step (in microns). Now, in your graphic, how many CFZ widths error are contained in the extreme spread (roughly 1000 steps) of the values you generated?
Thus, estimating a step position based on that graphic, is statistically more likely to produce an out-of-focus image, than one in focus.
Bart wrote: With elevation is meant the altitude of the object (as if using an alt-az mount), this -does- change during a session.
Also, if you take refraction into account, will you then -not- take it into account when using narrow band filters? There the effect isn't measurable.
Now a question, would you consider to focus during imaging or quickly adjust the focus in between lights?
Personally, I don't use very long exposure lights, especially with Starlink up and screwing up our images.
dmsummers wrote: @bdavis:
The refraction graph is about as I expected. The change from 40 to 25 degrees is on the order of an arcmin POSITION. Colors refract differentially, so we'd expect HFR to increase perpendicular to the horizon. Whether refraction or airmass is the major driver, I consider elevation to be a good proxy for modeling/controlling the effect (and it's readily available).
dmsummers wrote: I consider elevation to be a good proxy for modeling/controlling the effect (and it's readily available).
@dmsummers, I'm theorizing only, but given the data you presented, I'm going to bet on refraction being your main residual driver below 40 degrees elevation angle. My logic is as follows: 1. The focuser is mainly using HFR, which is a measure of the size of the circle encompassing the star in the image. 2. Refraction directly effects the measured size of the circle, mainly beginning at 40 degrees above the visual horizon, as illustrated below:
3. Changes in the HFR outside the expected due to step movement show up as residuals (error) in the measured HFR, forcing the measured points to fall outside the polynomial used to determine best focus.
4. Temperature, pressure, humidity. dust, in the air volume the light passes through, all change the refraction of received light
What I don't know, is the size of the HFR errors produced by refraction at 40 degrees and below, or their stability, or the extent to which the programmer factored it into to the focuser tools, but given the parameters you've listed, I'd be surprised if it wasn't the main driver.
Very interesting data. It would be interesting to map the two curves against the calculated refraction coefficient , since I think that's probably the largest part of the differentials you are seeing. Assuming a weather source is configured, Ekos should have that data somewhere, perhaps a script could be genned up to log the calculated R, temp, pressure, humidity, current step, elevation angle, and current HFR. This could be used by the focus module to maintain a quadratic that could be used to auto-configure the focuser at runtime, and perhaps condition other activities as well.
It works very well indeed, and my first tests with it have me slobbering at the improvements in guiding error. The old CGEM motor board and plastic spur gear could manage a 2.5 arcsec guide error RMS on a really good day. My initial tests with Onstep reduced that to around .75-1.0 arcsecs, and I haven't really tuned it yet, or connected the timing signal from the GPS to the oscillator on the Onstep board. I'm currently learning Fusion 360 and designing a case for it. Already printed the motor mounts for it, but had to back up and re-print them with a more temperature resistant plastic, which led to experimentation with annealing the plastic, etc.
Not sure about the CEM70 for Onstep, but the CGEM can be readily Onstepped with a remarkable improvement in accuracy and function. You can use mainboards that also support dew heater control, focuser control, and more. The whole rig can be customized in both software and hardware in a lot of ways, depending on your skill level.
I went a different way, and am upgrading the drive train and control electronics. The largest part of tracking error on a CGEM is the weak plastic spur gear, and cheap motors/motor control. The mount is stout otherwise. So replacing the old spur gear drive with modern stepper motors, and a 3d printer motor control board running OnStep software.
Onstep website is located HERE
Generally, the RPi4 has all 4 USB ports tied into a single PCIe lane limited to 4Gb/s (the SD card uses USB also). The whole Pi, has a max current limit of 1.2a, including board power, and whatever goes out the USB ports. This is why so many people have trouble running gear off the Pi's USB ports.
So, a good first step for problems with all USB gear attached directly to the Pi, is to move it off to a powered hub. Consider that any form of disk write, including images, temporary files, etc, are going to use a lot of USB bandwidth, and since USB 3.0 has a limit of 5Gb/s, and the Pi can do only 4, directly attached devices that use both current and bandwidth (like Cameras of any sort) are an iffy proposition at best, and often produce weird problems, problems caused by starving other devices, or just not working at all.